The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||20 September 2015|
|PDF File Size:||2.32 Mb|
|ePub File Size:||14.4 Mb|
|Price:||Free* [*Free Regsitration Required]|
To exit Erase mode without To erase all stored results: The CoaguChek XS PT test strip may be used for patients under a combination therapy of oral anticoagulants plus heparin injections. Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
Get Ready Take a test strip out of the then press M container.
Roche CoaguChek XS User Manual
Open Battery Compartment With the meter turned off, turn it over. Integrated Quality Controls Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing ,anual blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
Preparing For A Test 1. A rotation scheme was used to ensure that the same instrument was not always dosed first.
For the venous testing, blood was applied from a syringe; If the memory is result in memory. About this Manual Memory Cleaning the Meter It is important to keep the meter clean.
Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1. Gather Items Gather the following items: Keep the tube level, and allow it to fill by capillary action. Page 26 Repeat the test. The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips.
Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. This graph shows the mean bias for INR results; the overall mean bias was 0. Erasing Stored Test Results 2. Comments to this Manuals Your Name. Wurster M, Doran T. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples mabual very good.
Dose the Test Strip Discard the first 4 drops of blood. For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and manuxl second drop was applied to a second CoaguChek XS moni- tor. To print the manual completely, please, download it. Got it, continue to print. Record Result Record the result according to test strip. The correlation was 0.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
To go to of each letter, number, and symbol appear. Page 24 For more information, see re-insert it.
The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Performing A Test 3.
The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you are viewing a results and their dates and times in its memory.